In-Situ Oxide Scale Thickness Measurement
Super Heater and Re-Heater tubes carrying steam in fossil-fired boilers operating at temperatures above 9000 F (4820 C) are subject to failure by creep-rupture. These tubes form an internal oxide layer that inhibits heat transfer through the wall and causes the tube metal temperature to increase over time.
Ultrasonic testing provides a quick and non-destructive method for measuring scale. Hand-held, portable instruments such as the Olympus, 38 DL PLUS precision Thickness gauge with 20 MHz Shear wave Transducer can measure internal oxide layers down to a minimum thickness of approximately 0.150 mm.
38DL PLUS uses advanced algorithms to measure the thickness of oxide/scale buildups inside boiler tubes. The gage simultaneously displays the metal thickness of the boiler tube and the thickness of the oxide layer. Knowing the thickness of the oxide/scale helps predict tube life. We recommend using M2017 or M2091 transducers in this application.
WRITE TO US
Quality Network Pvt Ltd.
No. 10, II Cross Street, Kalaimagal Nagar,
Ekkattuthangal, Chennai - 600032
#38/GA, Shoreham, 5th Avenue, Besant Nagar,
Chennai – 600 090
Phone: +91 44 4857 5263
Mob: +91 9444979858
11 Main Street
Sparta, NJ 07871-1979
Phone: +01 (973) 726-8399
Mobile: +01 (973) 699-4460